Whether it’s a slip on the way to the mailbox or a fall while climbing a tree, accidents happen. Thankfully, titanium cochlear implants are generally robust against impact—but they can be dislodged if not thoroughly secured.1
Currently, there are several means of implant fixation in common use. Most often, a bony well is used to provide a stable bed for the receiver housing.2 This is usually combined with suturing across the implant for secure placement.1,2,3
However, a recessed implant bed can require significant drilling. This can expose the dura mater and increase the risk of complications—especially for children and patients with thinner temporal bone.2,4 Additionally, fixation of the stimulator housing with sutures requires a larger incision in order to expose the area surrounding the receiver bed. This can lead to increased scarring and risk of post-operative wound complication.1,3,4,5
Preparing a tight subperiosteal pocket allows you to minimize the size of the incision, avoids excessive drilling, and even eliminates the use of implant fixation sutures. However, reliance on a tight subperiosteal pocket alone can limit positioning options and fixation may be less secure.1 This is a significant consideration, especially for children, as they are prone to minor impacts to the head in everyday life.4,6
Therefore, an ideal fixation method would combine the best of both:
A) The long-term stability possible by drilling a bony well with fixation sutures.
B) The small incision and minimal drilling approach of a tight subperiosteal pocket.
Better Implant Fixation
In order to provide best of both, MED-EL developed the PIN implant housing. This design offers a reliable option for minimally invasive surgery and simple, secure fixation.1,3,4
The SYNCHRONY PIN Cochlear Implant offers reliable implant fixation with:
- Small incision behind the pinna (approx. 3 cm)
- Minimal drilling to flatten receiver area (<1 mm)
- No implant fixation sutures
SYNCHRONY PIN Cochlear Implant
SYNCHRONY PIN has two small (1.0x 1.4 mm) titanium fixation pins built into the receiver housing. These durable pins are designed to lock into corresponding holes drilled in the bone. This direct mechanical connection to the temporal bone provides superior fixation against lateral and rotational forces. Together with a sufficiently recessed electrode channel to protect the electrode lead, this provides excellent implant stability and security. Schnabl et al. (2012) found that even 100 N of force applied directly to the side of the PIN implant housing did not dislocate the implant.1
Furthermore, the area beneath the receiver housing only needs to be flattened. This minimizes the need for drilling and reduces the risk of exposing the dura mater, especially for patients with thin temporal bone.1,4
There’s no need for sutures to fixate the implant, so you won’t need to expose the area over the implant. This allows you to place the implant securely in a tight subperiosteal pocket.3,4 Only a small incision is needed behind the pinna. The small incision can reduce the risk of complications, enables faster wound healing, and minimizes visible scarring.3,4,5,6
This simplifies surgery for you and offers better outcomes for your patients.1,2,3,4,5,6,7
The benefits of PIN for you:
- Simple, reliable implant fixation 3,4
- Streamlined surgical procedure 3,4
- Reduced surgical time 3,4
- Less risk of contact with dura mater 2,3
- Low post-operative complication rate 3,4
The benefits of PIN for your patients:
- Impact-resistant stability for children 1,4
- Superior fixation for long-term reliability 1,3
- Less risk of surgical complications 2,3,4
- Faster healing, fewer complications, and less scarring 2,3,4,5
With straightforward implant fixation, SYNCHRONY PIN offers surgical simplicity, faster recovery, and long-term reliability.3,4,7
Want to start using SYNCHRONY PIN in your clinic? Contact your local MED-EL representative today.
Don’t miss the next post—subscribe now and you’ll receive new updates right in your inbox.
*Not all products and features shown are available in all areas. Please contact your local MED-EL representative for more information.
- Schnabl, J., Markl, A., Hoermann, R., Wolf-Magele, A., Schartinger, V., & Sprinzl, G. (2012) Concerto Pin: a novel concept of cochlear implant fixation. Otol Neurotol. 33(9): 1525–1529.
- Yoshikawa, N., Hirsch, B., & Telischi, F.F. (2010) Cochlear implant fixation and dura exposure. Otol. Neurotol. 31(8): 1211–1214.
- Kuzovkov, V., Sugarova, S., & Yanov, Y. (2016) The Mi1000 CONCERTO PIN cochlear implant: An evaluation of its safety and stability in adults and children. Acta Otolaryngol. 136(3):236–240.
- Schnabl, J., Wolf-Magele, A., Pok, S.M., Url, C., Zorowka, P., & Sprinzl, G. (2015) Evaluation of a minimally invasive surgical fixation technique for young children with the Concerto Pin cochlear implant system. Eur Arch Otorhinolaryngol. 272(8): 1893–1898.
- Cui, D., Shi, Y., Su, Q., Liu, T., Han, D., & Li, Y. (2014). Minimal incision access for pediatric and adult cochlear implantation. Chin Med J (Engl.) 127(13):2434–2437.
- Davids, T., Ramsden, J.D., Gordon, K.A., James, A.L., & Papsin, B.C. (2009) Soft tissue complications after small incision pediatric cochlear implantation. Laryngoscope. 119(5): 980–983.
- MED-EL Cochlear Implant Reliability (October 2016) http://www.medel.com/int/reliability-reporting