People with Congenital Microtia-Atresia (CMA) have malformations of the pinna and the external auditory canal. Often the middle ear anatomy is also affected and shows abnormalities. The vast [...]
MRI is an essential tool in modern medicine. Our priority has always been to offer safe, comfortable access to MRI scans for our cochlear implant recipients. With a secure magnet design, there [...]
Why did we create FLEXSoft, the longest cochlear implant electrode array available? The simple answer is that we wanted to provide an array that was long enough to stimulate as much of the [...]
Enhanced 3D anatomy, individualized audiogram & electrode visualization, virtual trajectory planning and more—check out the latest updates to OTOPLAN!
Incredibly detailed 3D models of inner ear, facial nerve, and more: Don't miss this fascinating article from Daniel Sieber, Head of MED-EL’s Electrodes and Surgical Research & Development.
Watch now: In this HD video, PD Dr. med Henning Frenzel shares step-by-step guidance on his surgical techniques for the VIBRANT SOUNDBRIDGE Middle Ear Implant.
Why does tip fold-over happen with certain electrode arrays? See why you can rely on MED-EL arrays in this overview of recent research covering more than 2,000 cochlear implant cases.
The dawn of a new era in cochlear implant surgery: Prof. Vedat Topsakal & Prof. Paul Van de Heyning share their surgical experience with the first HEARO robotic cochlear implantation procedure at [...]
In this HD surgical video, Dr. Harold Pillsbury guides us through his techniques for implanting a titanium MED-EL cochlear implant with a full-length electrode array.